The tester features a unique metal surface electrode that provides fast and easy on the spot EC(mS/cm) measurements for soil and hydroponics nutrient etc.
- Take off the protective cap from metal probe
- Press OFF/ON button Tester display “0″.
- Quick testing : Water the soil with distilled water before testing. Make sure the water can go 10 cm deep the least. Best humidity is 60%-80%. Insert the metal probe into soil vertically and clockwisely, to about 6-8 cm deep. Make sure the soil have thorough contact with metal probe. After about 6 seconds, the tester will display the measured value. Tightness of soil will affect measuring result. Multiple tests at the same and different points should be performed and take the average as final result.
Accurate testing : Use a container, mixdried soil with distilled water at the rate of 10:1. Stir and make it into a mud form. Put the metal probe into the mud about 6-8CM deep. About 6 seconds, the tester will display the measured value.
- Wipe the metal probe with towel or tissue paper after each test. Make it clean and dry. till the display show “0″.
Technical Specification (6 Position):
|Range||: 0-1999 EC(mS/cm);|
|Resolution||: 0.10 EC(mS/cm)|
|Accuracy||: ± 0.1EC (00-100 EC)|
|Operating Temp||: 5-45°C (41 – 113°FVATC|
|Calibration||: 1 point (6.7EC /Factory calibrated -No more calibration)|
|Power supply||: DC 9V(L1022 battery)|
|Display||: LCD with illumination|
|Dimension||: Æ50.5 x 250 mm|
|Probe length||: 140mm|
|Features||: Automatic switch-off function : Aer about 5 minutes without pushing
buttons, the meter will be switched off automatically (or manual shutdown)